- SEE ALSO
- COPYRIGHT AND LICENSE
Lab::Instrument::OI_ILM210 - Oxford Instruments ILM Intelligent Level Meter
use Lab::Instrument::OI_ILM210; my $ilm=new Lab::Instrument::OI_ILM210( connection_type=>'IsoBus', base_connection=>..., isobus_address=>5, );
The Lab::Instrument::OI_ILM210 class implements an interface to the Oxford Instruments ILM helium level meter (tested with the ILM210).
my $ilm=new Lab::Instrument::OI_ILM210( connection_type=>'IsoBus', base_connection=> $iso, isobus_address=> $addr, );
Instantiates a new ILM210 object, attached to the GPIB or RS232 connection (of type
$iso, with IsoBus address
Reads out the current helium level in percent. Note that this command does NOT trigger a measurement, but only reads out the last value measured by the ILM. This means that in slow mode values may remain constant for several minutes.
As optional parameter a channel number can be provided. This defaults to 1.
This software is copyright (c) 2017 by the Lab::Measurement team; in detail:
Copyright 2011 Andreas K. Huettel, Florian Olbrich 2012-2013 Andreas K. Huettel 2016 Simon Reinhardt 2017 Andreas K. Huettel
This is free software; you can redistribute it and/or modify it under the same terms as the Perl 5 programming language system itself.